- You must be logged in to reply to this topic.
I am trying to calibrate Stark 9-66A tube tester and have a problem. Instructions specify to measure voltage of 20 VAC when P4 button is pushed and 0 VAC when P6 is pushed. I have the opposite readings, 0V when P4 is pushed and close to 20 VAC when P6 is pushed.Can anyone help? DC readings at P4 are close to 220V as specified by instructions. Calibration instructions are attached.Thanks in advance.
Click on the different category headings to find out more. You can also change some of your preferences. Note that blocking some types of cookies may impact your experience on our websites and the services we are able to offer.
These cookies are strictly necessary to provide you with services available through our website and to use some of its features.
We provide you with a list of stored cookies on your computer in our domain so you can check what we stored. Due to security reasons we are not able to show or modify cookies from other domains. You can check these in your browser security settings.
These cookies collect information that is used either in aggregate form to help us understand how our website is being used or how effective our marketing campaigns are, or to help us customize our website and application for you in order to enhance your experience.
If you do not want that we track your visist to our site you can disable tracking in your browser here:
We also use different external services like Google Webfonts, Google Maps, and external Video providers. Since these providers may collect personal data like your IP address we allow you to block them here. Please be aware that this might heavily reduce the functionality and appearance of our site. Changes will take effect once you reload the page.
Google Webfont Settings:
Google Map Settings:
Vimeo and Youtube video embeds: